The cyclone can be disassembled by the user for cleaning. After cleaning, we recommend a leak test of the system using the FSP10/FSP10-2 silicone adapter (Art.No.: 00161) The coarse dust container (Gritt pot) must be emptied after each sampling, as the coarse dust fraction collects in it.
The universal holder is screwed onto the thread of the cyclone.
The filter holder is fitted with a membrane filter with a pore size of 8 µm and an outer diameter of 37 mm. In contrast to the GSP, GGP and FAP systems, with the FSP the inflow to the filter is from below through the universal holder. Accordingly, the filter side to be loaded must be inserted into the filter holder from below. The support strainer must be placed between the filter and the clamping ring. The suction funnel is placed concentrically on the clamping ring and fixed in the universal receptacle with the union nut. A flat gasket is inserted between the suction funnel and the filter capsule.
The FSP10-2 sampling head complete consists of:
PGP universal receptacle, FSP10/FSP10-2 suction funnel, FSP flat gasket, FSP10-2 cyclone, coarse dust container (Gritt pot), three spare flat gaskets.
Upon request, we can provide you with a cleaning recommendation.